This is a simple visible light meter that is very easy to build, as well as inexpensive (around $5 for parts from DigiKey). The IS474 is a visible light sensor that has the very useful feature of passing .1 uA per lux, which can easily be converted to a measurable voltage using standard value resistors. There is a three position switch in this schematic, which allows you to choose the measurement level for better resolution, with a maximum reading of 35,000 lux. Then, the voltage across that resistor is simply measured with a digital volt meter - inexpensive ones are available at Wal-Mart or Radio Shack for under $20. The voltage output is 0 - 3.50 volts, in which the conversion to lux varies depending on which position the sensitivity switch is set on. Also note that this voltage could also be read on an Analog to Digital Converter, but it would be necessary to buffer the output impedance with a voltage follower after the + terminal of Vout. The IS474 can be mounted in many ways - as long the light "window" is exposed to whatever light source you want to measure. The power supply can be from a power transformer, 9 volt battery, or any other 7-25 volt source.
For my meter, I modified the schematic a bit - instead of using a voltmeter to read the output, I used an ammeter (0 - 50 uA) to measure the current from the IS474 directly. For the sensitivity switch, the first position just measures the entire current from the IS474, and for the other two positions I used a current divider relationship to put 1/10th and 1/100th the current through the ammeter. This was a bit tricky to calculate, and the resistor values will depend on the internal resistance of the ammeter, but the extra effort was worth it. Below is a photo of my visible light meter - built into a nice handheld package, with an analog meter face, along with the spectral response of the IS474 (which is very similar to that of the human eye):
<- Back to Main How-To Page
All materials Copyright 2002 - 2009 by JDF of GhostGadgets.com
All Rights Reserved